|Nr 1, Tech 9th Road, Strefa rozwoju High & New Tech, Songshan Lake, DongGuan City, Prowincja Guangdongfirstname.lastname@example.org|
|Control Box:||PLC Touch Screen Control||Standards:||EN 1728:2012; BS EN 581-2|
|Purpose:||Seat and back Static load test|
TNJ-017 Seat Endurance Tester--Office Chair Seat and Backrest Strength and Durability Tester
Purpose: This test apparatus is used for the strength and durability performance test for office chairs, including seating load test, backrest load test and seat & back combined test.
Principle: This test apparatus do the durability test to chair seat and backrest by simulating the loading application to the chair as daily use and habit misuse. First use the seat loading pad to apply the certain loading to the seat surface, then the back loading pad to the backrest, then loading unload to finish one cycle of test, repeat the loading movement some times and certain cycles according to different test requirement, to determine the durability performance of chairs.
Office Chair Seat Back durability tester is used for testing the strength and durability of office chair. Made of Aluminum material frame, PLC control, easy operation; high precision force sensor for the load application.
Seat and chair back durability combined tester.
Seat and back Static load test: test consisting of heavy loads being applied a few times to ensure that furniture has sufficient strength under the highest levels of loading that might reasonably be expected to occur.
Seat and back Fatigue test: Test simulating the repeated application of loads or movement of components occurring during long-term functional use.
This test apparatus do the durability test to chair seat and backrest by simulating the loading application to the chair as daily use. First use the seat loading pad to apply the certain loading to the seat surface, then the back loading pad to the backrest, then loading unload to finish one cycle of test, repeat the loading movement some times and certain cycles according to different test requirement, to determine the durability performance of chairs.
Pneumatic drive to load, easy to clean.
Loading Sensor: 4000N (1nos), 1000N (1nos)
Loading force for seat testing: 0-4000N (adjustable)
Loading force for back testing: 0-1000N (adjustable)
Load force solution: 1/10000
System test force accuracy: ±1%(static) ; ±5% (dynamic)
Test speed:10~30 cycles per minute, 1-5 test cycles per 10 second (adjustable)
Loading cycle:0~999999 adjustable
Control method:Microcomputer automatism servo control, Touch screen control
Loading method: Pneumatic linear actuator
Loading power: Motor
Specimen test station: 1
Test independence: Independent control by each work station
Setting: Computer software setting
Data storage: RAM and U disc
Test report: Make the report on computer as request
Loading force adjustment:Air pressure adjustment (300N-3500N)
Motion control:PLC control
Backrest loading height: Manual adjustment
Air source:Air supply input is 0.4Mpa-0.8Mpa. (6kgf/cm^2 or above, supplied by customer themselves)
Power:1∮AC 220V-240V 50Hz 5A, 3KW
Accessories:Seat loading pad 1 set; back loading pad 1set; Stationary fixture 1 set.
Standards: EN 1728:2012 6.4 Seat static load and back static load test; EN 1728:2012 6.17 Combined seat and back durability test; BS EN 581-2 B.2.1.1 and B.2.1.2,
Apparatus structure: This tester uses aluminum alloy frame, iron and aluminum plates for the structure manufacturing. Motor part has a air cylinder operation with electronic components for operation.
Machine dimension: Length 1400mm x wide 1400mm x height 2000mm
Control box dimension: Length 640mm x wide 600mm x height 850mm
Spare Parts:1. Distribution box with aluminum alloy frame x 1
2. Portable magnetic sucker x 4
3. Loading pad x 2
Osoba kontaktowa: tonyhkhk